The electrodes are designed for physiotherapy procedures. The electrodes can be used in the medium with pH value ranging 3-10 at +10 - +40ºC. The electrodes shall be stored at -50 - +50C and relative humidity not exceeding 80%. Each electrode has eight bleached flannel layers and one conductive graphitized layer; the electrodes are connected by conductor pad of flag type. The electrode should fit tight against the contact surface. The conductor pad should be well-fixed to the active electrode surface. Before application, the electrodes shall be well moistened and slightly squeezed, then fixed on the patient’s body by rubber bandages and tourniquets or bags with sand. The electrode should fit tight against the body with the entire surface. Connect the conductor pad to the active electrode surface. After application, the electrodes shall be sterilized by boiling in distilled water for 30 min. Electrode’s average lifetime is 30 cycles of disinfection in distilled water according to Sanitary Regulations OST 42-21-2.